# A fast image simulation algorithm for scanning transmission electron microscopy

- Colin Ophus
^{1}Email authorView ORCID ID profile

**3**:13

https://doi.org/10.1186/s40679-017-0046-1

© The Author(s) 2017

**Received: **2 March 2017

**Accepted: **27 April 2017

**Published: **10 May 2017

## Abstract

Image simulation for scanning transmission electron microscopy at atomic resolution for samples with realistic dimensions can require very large computation times using existing simulation algorithms. We present a new algorithm named PRISM that combines features of the two most commonly used algorithms, namely the Bloch wave and multislice methods. PRISM uses a Fourier interpolation factor *f* that has typical values of 4–20 for atomic resolution simulations. We show that in many cases PRISM can provide a speedup that scales with *f*
^{4} compared to multislice simulations, with a negligible loss of accuracy. We demonstrate the usefulness of this method with large-scale scanning transmission electron microscopy image simulations of a crystalline nanoparticle on an amorphous carbon substrate.

## Keywords

## Background

Transmission electron microscopy (TEM) is one of the most versatile and powerful experimental tools for imaging and diffraction of micrometer to sub-nanometer structures. The recent widespread adoption of hardware aberration correction has, in particular, enabled routine atomic resolution imaging of structures [1–3]. A more recent technical advance for TEM experiments is the use of direct electron detectors. These cameras have a much higher quantum efficiency than standard charge-coupled devices with a scintillator, and can also operate at much higher speeds [4–7]. Direct electron detectors have already created dramatic improvements in plane wave TEM imaging experiments, especially single-particle biological cryo-EM studies [8–10]. These detectors have also enabled many new kinds of experiments for scanning transmission electron microscopy (STEM), where the electron probe is converged to very small dimensions and scanned across the surface of a sample, because the camera speed is high enough to record a full image of the diffracted probe at each probe position [11]. Examples include nanobeam electron diffraction strain measurements [12, 13], orientation mapping of semi-crystalline polymers [14], and phase contrast imaging modes such as differential phase contrast [7, 15], phase plate interferometry [16], and ptychography [17]. Each of these experiments can benefit by accompanying them with STEM simulations to aid in interpretation or validation of the results.

However, while the experimental capabilities of TEM and STEM have expanded, simulation methods have remained largely unchanged for some time. The two primary methods currently used for atomic resolution simulations are Bloch wave calculations and the multislice method [18–20]. In the Bloch wave method, the electron wavefunction is defined using a basis set that satisfies the Schrödinger equation inside the sample. For a perfect crystal, Bloch waves are stationary solutions with the same periodicity and symmetry as the crystalline lattice. After calculating the eigenvectors and eigenvalues of this basis set, the wavefunction at the entrance surface of the sample can be matched to the known electron probe coefficients, and then the resulting electron wavefunction can be computed everywhere (including the exit surface of the sample) [21, 22]. This scattering calculation can be written compactly in a scattering matrix (often called the **S**-matrix) formalism [20]. Bloch wave calculations are almost never used for imaging or diffraction simulations of large samples (beyond the several ‘unit cell’ scales for crystalline materials) for two reasons; the first is that eigendecomposition of a non-sparse Bloch wave matrix large enough to accurately simulate image sizes ≥1000^{2} pixels would take an impractically long time to compute. The second is that the storage requirements of this scale of **S**-matrix are greater than a terabyte, and using it would require trillions of multiplication operations [20].

A more efficient formulation for large electron scattering simulations is the multislice algorithm [23]. In this method, the atoms of the simulated sample are divided up into infinitely thin slices along the beam direction. The resulting electron scattering is calculated by alternating between a transmission operator through each slice, followed by Fresnel propagation of the electron wave to the next slice. These operations can be performed efficiently in realspace and reciprocal space, respectively, and so an efficient implementation of this method requires a forward and inverse Fourier transform at each step [20]. The multislice algorithm is very efficient for plane wave, conventional TEM image, or diffraction simulations. It is much less efficient for STEM simulations consisting of thousands or millions of probe positions. This is because while the atomic scattering potential can be reused for all probe positions, the transmission and propagation steps must be repeated for each additional probe position. The scattering potential calculations can be performed very efficiently using lookup tables [16, 24] or a point scattering method [25], but the slow part of the calculation is usually repeated for all probe positions [20]. Many STEM studies such as high precision 2D measurements [26–28], 3D atomic electron tomography [29, 30], and others [31] make use of image simulations of many thousands of STEM probe positions. This requires long computation times, even with modern implementations of the multislice method [25, 32–37]. It is therefore desirable to develop an electron scattering simulation algorithm that shares the calculation burden between STEM probe positions in a more efficient manner than multislice simulation. Chen et al. [38] have proposed one such method, but it has not found widespread application. For a detailed discussion of the relationship between the Bloch wave and multislice simulation methods, we refer readers to the derivations of Allen, Findlay et al. [18, 19].

In this manuscript, we derive a more efficient algorithm for STEM simulations by combining aspects of the multislice and Bloch wave methods. We use the multislice method to directly calculate a subset of the rows of the **S**-matrix (corresponding to plane waves of various orientations), which is then used in a similar manner as Bloch wave calculations [38] to relate the output wavefunction to a given input. The key insight is that because highly converged STEM probes decay to zero quickly with distance from the probe center position, they can be cropped out of the full **S**-matrix in a highly accurate Fourier interpolation scheme. The algorithm presented here is referred to as the plane wave reciprocal space interpolated scattering matrix (PRISM) algorithm. We also compare the accuracy and computation time of the PRISM and multislice algorithms, and suggest some useful extensions of the PRISM method.

## Theory and methods

### The multislice and Bloch wave methods

*z*can be described by the Schrödinger equation for fast electrons [20]

*λ*is the relativistic electron wavelength, \({\nabla _{xy}}^2\) is the 2D Laplacian operator,

*σ*is the relativistic beam sample interaction constant, and \(V (\vec {r})\) is the electrostatic potential of the sample.

**S**as [20]

**S**-matrix may contain billions or more entries, which requires an impractical amount of time to calculate the eigendecomposition. And, actually using Eq. 2 many times for various electron probes could take a very long time. Thus, Bloch wave methods are typically only used for very small size STEM simulations.

*t*taken from the sample. The left term is interpreted as a Fresnel propagation operator, which can be efficiently applied in Fourier space as [20]

*p*refers to the slice index. The second operator of Eq. 1 can be efficiently applied in real space as

*p*, integrated over the slice along the beam direction from the 3D potential. In practice, the atomic potentials are integrated into 2D potentials before the simulation, and then added directly to the slice potential, or applied using convolution [25]. These two steps describe how the electron wavefunction evolves slice-by-slice until it has interacted with the entire sample, applied sequentially as

**S**-matrix approach, where the computational load of applying Eq. 5 can be shared between different probe configurations.

### The PRISM algorithm for STEM simulations

The first step of the method proposed here is to separate all atomic coordinates of the simulation cell (which is assumed to be orthorhombic here) into slices, as shown in Fig. 1a. These slices can have unequal thickness to better match the atomic coordinates, but should not have thicknesses larger than the average atomic spacing as this could cause errors [20]. The second step is to calculate the 2D projected potentials \(V(\vec {r})\) for all slices, as shown in Fig. 1b.

*f*. In practice, a different factor can be used in

*x*and

*y*, but for simplicity we will describe the simulation method for a square [in the (

*x*,

*y*) plane] simulation cell of size

*d*. This factor

*f*should be chosen to be large enough so that a square area with a side length of the simulation cell size divided by

*f*can encompass all possible STEM probes after they pass through the cell. This can be estimated by numerically simulating a few probes using the conventional multislice method or the method described here. We then also choose a maximum incident probe semi-angle \(\alpha _{\mathrm{max}}\). Note that the simulation will include larger scattering angles than this value, and that this value should be equal to the largest desired probe semi-angle plus

*f*times the Fourier space pixel size \(\Delta q\). We then determine a set of plane wave initial conditions to simulate using the multislice method, as shown in Fig. 1c. This set of plane waves corresponds to the incident electron probe

*m*,

*n*) are integers representing the plane wave index. Thus, we compute only a subset of all possible periodic plane waves for the simulation cell size, reducing the number of waves calculated by a factor of

*f*

^{2}. These plane waves are stored in realspace in a large array that we will refer to as the compact

**S**-matrix, with the output plane waves defined as \(\mathbf {S}_{m,n}(\vec {r})\). These output wave dimensions can be reduced by a factor of 4, if the multislice simulation uses an antialiasing aperture positioned at half of the maximum scattering angle [20].

*d*centered around the probe. This is shown schematically in Fig. 1d. The subregion is bounded by

*d*

^{2}/

*f*

^{2}, which should be periodically wrapped around the simulation cell boundaries. The wave coefficients are defined as

*C*

_{1}or 3rd order spherical aberration

*C*

_{3}described by the phase shift function [20]

*h*and probe tilt angles \(\theta _x\) and \(\theta _y\). As shown in Fig. 1e, once the probe coefficients \(\alpha _{m,n}(\vec {r}_0)\) have been computed, the complex probe in realspace \(\psi (\vec {r},\vec {r}_0)\) can be computed using the summation

### Simulation and analysis implementation

All simulations and analysis in this study were performed using custom Matlab code. The multislice methods and the atomic potentials employed were taken from Kirkland [20]. Thermal scattering effects were implemented using the frozen phonon approximation, which involves repeating the calculation with different phonon configurations (approximated with random atomic displacements) and summing the results incoherently.

An implementation of the PRISM algorithm for a sample consisting of a nanoparticle contained within a carbon nanotube is shown in Fig. 2a–f. Each of the panels in this figure corresponds to the same step as those given in Fig. 1a–f. In Fig. 2c, e and f, the wave phase is shown as the color hue, while the wave amplitude is shown by the brightness of each pixel. All simulations were performed using an 80 kV accelerating voltage, a slice thickness of 0.2 nm, a pixel size of 0.01 nm, and we used no spherical aberration in the electron probes.

### Calculation time for PRISM simulations

*N*where

*N*is a power of two. Each slice will require the transmission and propagation operations given in Eq. 5, which requires \(6 N \log _2(N)\) complex operations for the forward and inverse Fourier transforms and 2

*N*

^{2}operations to multiply the sample potential and the Fresnel propagation functions. If the entire STEM simulation consists of

*P*unique probe positions and

*H*slices through the sample, the total calculation time \(T_{\mathrm{multi}}\) required is

*B*/

*F*

^{2}multislice simulations, where

*B*is the number of beams included in the full-resolution simulation, which will be reduced by the interpolation factor squared. The second half is the multiplication of the compact scattering matrix

*S*for all beams (multislice plane waves computed in the previous step), which is required for

*P*total probes, as in Eq. 10. This multiplication step is only required for the reduced number of beams

*B*/

*F*

^{2}, and the cut out region defined by Eq. 7 will reduce the number of multiplication operations to

*N*

^{2}/4

*f*

^{2}(note the extra factor of 1/4 is due to storing only the part of

*S*inside the anti-aliasing aperture). Therefore, the total calculation time \(T_{\mathrm{PRISM}}\) required for PRISM is

*B*used in practice is often much lower than the number of possible beams. The speedup offered by the PRISM algorithm is therefore approximately equal to the ratio of Eqs. 11 and 12 given by

*S*-matrix, the speedup ratio does not depend on the number of probe positions

*P*and the speedup will vary with

*f*

^{4}. In the multislice and PRISM simulations given in the first results section below, the values of the terms of Eq. 13 were

*H*= 40,

*B*= 10

^{4}, and

*P*= 10

^{5}. Plugging these numbers into Eq. 13 gives a speedup factor \(T_{\mathrm{Multi}} / T_{\mathrm{PRISM}}\) of approximately 0.5, 8, 110, and 1100 for

*f*= 2, 4, 8, and 16, respectively.

## Results and discussion

### Comparison of accuracy between multislice and PRISM

In general, PRISM will always be less accurate than corresponding multislice calculations, unless the PRISM speedup allows for finer pixel sampling, inclusion of higher scattering angles, or a similar improvement. However, the increased error is negligibly small in many cases, and will depend heavily on the microscope and sample parameters of a given simulation. To demonstrate this, we have compared the accuracy of a STEM probe calculation for a typical experimental geometry: a Pt nanoparticle (NP) approximately 7 nm diameter tilted 30° from the primary axis. The NP rests upon an amorphous carbon substrate with a thickness of 5 nm, as shown in Fig. 3a. The NP has a multiply twinned decahedral structure, with screw and edge dislocations present in two of the grains. The NP atomic coordinates were taken from [39], and the amorphous carbon structure was adapted from [40].

The sample was divided up into slices 0.2 nm thick, and the projected potential was computed for all slices. The sum of these potentials is shown in Fig. 3b, with an enlarged inset shown in Fig. 3d. The initial STEM probe generated from a 25 mrads semi-angle aperture at 80 kV is shown in Fig. 3c, with the probe center position shown in Fig. 3d. We then calculated the probe wavefunction after passing through the sample using the multislice method (Fig. 3e) and the PRISM algorithm with interpolation factors of \(f=\) 2, 4, 8, and 16 (Fig. 3f–i). The corresponding probe amplitudes in Fourier space are shown in Fig. 3j–n, respectively, and the logarithm of the radially integrated intensities is plotted in Fig. 3o–s, respectively. In the real space images, the channeling effect along aligned atomic columns is visible in all simulations [41].

We see that the PRISM method correctly reproduces most of the fine structure in the real space probe images. In Fig. 3i, we see that when \(f=16,\) the tails of the probe have been cut off by the edge of the cropping window, leading to small artifacts at the boundary (shown with white arrows). However, Fig. 3n, s shows that this simulation can still qualitatively reproduce the diffracted probe signal with good accuracy.

*f*increases in Fig. 3k–n. This is reflected in the radially integrated intensities, as a small mixing between adjacent detector angle bins. Secondly, there is a small decrease in intensity at the highest scattering angles. This decrease is very small, visible only because of the logarithmic intensity scale. The source is probably the interpolation step of PRISM, which will reduce the image sharpness slightly, manifesting at the highest spatial frequencies/scattering angles. We therefore conclude that PRISM is accurate enough to replace the traditional multislice method for STEM simulations in most cases. The primary exceptions are when the probe is very large (highly defocused or delocalized) or when fine details must be recovered from diffraction pattern, such as higher order Laue zone line measurements [42].

To demonstrate the accuracy of PRISM, we have performed full-image simulations of the sample shown in Fig. 3. Simulated STEM images are shown for various virtual detectors using the multislice method in Fig. 4a. PRISM simulations using interpolation factors of \(f=5\), 10, and 20 are plotted in Fig. 4b–d, respectively. The simulations correspond to cut out regions with side length 2, 1, and 0.5 nm, respectively. Note that the annular detector inner angle in the third row of simulations in Fig. 4 is slightly increased to prevent sampling artifacts at the edge of the 20 mrad semi-angle electron probe.

Figure 4a, b demonstrates that for relatively low interpolation factors, PRISM is essentially identical to Multislice simulations. PRISM can accurately capture the coherent diffraction image contrast present at lower scattering angles. Additionally, it can reproduce the clean mass-thickness contrast signal present at high scattering angles. As the interpolation factor is increased, subtle differences from the multislice image simulations do emerge, in Fig 4c–d. However, the image contrast is still qualitatively very similar to the multislice images. The primary advantage of PRISM is the reduced calculation time; the PRISM simulations with interpolation factors of \(f=5\), 10, and 20 gave speed up factors of approximately 40, 280, and 2100, respectively, compared to the multislice simulation. The \(f=20\) simulation shown in Fig. 4d requires only a few minutes of calculation time on a modern desktop computer, using Matlab code that has not been highly optimized or compiled.

Figure 4e shows an error estimate for the three PRISM simulations in Fig. 4b–d. The error was estimated as \(1-R^2\), where \(R^2\) is the correlation coefficient between the multislice and PRISM simulation pixel intensities. The \(f=5\) PRISM simulation error is approximately 0.005% for all scattering angles, indicating that this simulation is essentially error-free. When the interpolation factor is increased to \(f=10\), the difference from a multislice simulation increases to an error of 0.05% for low scattering angles and ≈1% error for intermediate scattering angles, and finally \(\approx\)10% for high scattering angles. Doubling the interpolation factor again to \(f=20\) gives 1% error at small scattering angles and 10% error for medium and high scattering angles. This larger error is caused by the region cropped around the STEM probe being small enough to crop out a significant portion of the probe intensity and cause boundary errors, as shown in Fig. 3i. We conclude that when using a low enough interpolation factor *f*, the PRISM method can simulate STEM intensities at all scattering angles with negligibly low error. The best value for *f* can be determined by testing probes at different locations in the simulation cell with both PRISM and multislice, or by using a conservative, low value; for example, in this simulation \(f=5\) leads to a cutout region with side length 2 nm, large enough to contain the entire STEM probe for any probe semi-angle large enough to generate atomic resolution contrast.

Figure 4f–h shows the difference in intensities between the PRISM image simulations in Fig. 4b–d, respectively, and the multislice image simulations in Fig. 4a. The intensity range for each panel is set individually to show good contrast for the features present. Figure 4f shows that when *f* is small, PRISM will slightly over-estimate the image intensity at scattering angles below the probe semi-angle, and slightly under-estimates the intensity at higher scattering angles. These intensity differences are probably caused by the different sampling of PRISM compared to multislice for both defining the initial electron probe, and creating the virtual detectors for the output signal. The errors for \(f=5\) also appear to be primarily intensity errors, which will not strongly affect measurements such as peak position estimation. Fig. 4g and h shows larger intensity differences for \(f=10\) and 20. These differences depend on the amount of local scattering and the local tilts of atomic columns, which could introduce errors in peak position measurements.

To test the accuracy of using PRISM to estimate structural information, we have used non-linear least squares peak fits using a 2D Gaussian function to measure 360 of the strongest peaks on the right-hand side of the decahedral particle as shown in Fig. 4. These peaks were measured from low-angle annular dark field (LAADF) images created with a virtual detector from 22.5 to 105 mrads. We have plotted the differences in measured peak positions between multislice and PRISM simulations in Fig. 4i–k, and the peak intensity differences in Fig. 4l–m. The mean 2D position errors for the PRISM simulations are 0.86, 2.8, and 21 pm for \(f=5\), 10, and 20, respectively. These errors will decrease if more frozen phonon configurations are included due to the increasing smoothness of the peak functions in the simulated images. Additionally, the errors could be reduced by using a probe sampling finer than 0.25 Å. The intensity differences in the peaks are fairly small, ≈1% for both \(f=5\) and 10. When *f* is increased to 20, the intensity errors increase rapidly, underlining the importance of choosing an *f* value low enough for the desired accuracy.

### PRISM simulations with varying probe size

In the PRISM method, once the compact *S*-matrix is computed for a given set of atomic coordinates, it can be used for many different simulations. The primary change between electron probes is not only the probe center position, but we can also vary coherent wave aberrations in the probe such as defocus or spherical aberration, change the probe size by modifying the probe semi-angle radius, and also include relative tilt between the probe and sample by moving the probe center away from \(\vec {q}=(0,0)\). These simulation parameter changes reflect only changes in the probe weighting coefficients \(\alpha _{m,n}(\vec {r}_0)\), given in Eq. 10.

*S*-matrix to simulate STEM images with varying probe size and annular detectors is shown in Fig. 5, for an accelerating voltage of 80 kV and a probe spacing of 0.025 nm. Based on the previous results shown in Fig. 4, we have chosen an interpolation factor of \(f=5\) for these simulations. In Fig. 5, we have generated annular bright-field images by setting the detector inner and outer angles to \(\approx\)75 and 100% of the probe semi-angle, respectively. Annular dark field images were generated by setting the detector inner angle to 40 mrads outside of the probe semi-angle. These simulations show that for this sample, a 10 mrad probe does not generate atomic resolution contrast. However, a 20 mrad probe can resolve the atomic columns of the two grains on the right-hand side of Fig. 5. Resolving atomic columns over the entire particle requires increasing the probe semi-angle to 40 mrads.

## Conclusion

In summary, we have presented the PRISM algorithm for STEM image simulation, which combines aspects of the Bloch wave and multislice simulation methods. PRISM uses Fourier interpolation with an integer factor *f* and can lead to a decrease in computation time that is proportional to \(f^4\) in many cases. We have compared PRISM and multislice image simulations and shown that as long as *f* is kept small enough, the simulation error for PRISM is negligibly small. Large *f* values can be used to generate a rough contrast model for a given simulation cell in very short computation times. We expect that the PRISM method will find wide application in STEM studies that require image simulation, due to its potential for a large speed up relative to the multislice method.

## Declarations

### Acknowledgements

We thank Earl Kirkland, Christoph Koch, and Roar Kilaas for their helpful discussions on (S)TEM simulation methods. We also thank Hao Yang, Jim Ciston, Tyler Harvey, and Peter Ercius for their helpful suggestions on this manuscript.

### Competing interests

The author declares that he has no competing interests.

### Availability of data and materials

Please contact the author for code examples and updates.

### Consent for publication

I consent for this manuscript to be published under the Creative Commons Attribution 4.0 International License.

### Funding

Work at the Molecular Foundry was supported by the Office of Science, Office of Basic Energy Sciences, of the U.S. Department of Energy under Contract No. DE-AC02-05CH11231.

### Publisher's Note

Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.

**Open Access**This article is distributed under the terms of the Creative Commons Attribution 4.0 International License (http://creativecommons.org/licenses/by/4.0/), which permits unrestricted use, distribution, and reproduction in any medium, provided you give appropriate credit to the original author(s) and the source, provide a link to the Creative Commons license, and indicate if changes were made.

## Authors’ Affiliations

## References

- Batson, P., Dellby, N., Krivanek, O.: Sub-ångstrom resolution using aberration corrected electron optics. Nature
**418**(6898), 617–620 (2002)View ArticleGoogle Scholar - Rose, H.: Prospects for aberration-free electron microscopy. Ultramicroscopy
**103**(1), 1–6 (2005)View ArticleGoogle Scholar - Dahmen, U., Erni, R., Radmilovic, V., Ksielowski, C., Rossell, M.-D., Denes, P.: Background, status and future of the transmission electron aberration-corrected microscope project. Philos. Trans. Royal Soc. Lond. A Math. Phys. Eng. Sci.
**367**(1903), 3795–3808 (2009)View ArticleGoogle Scholar - McMullan, G., Faruqi, A., Clare, D., Henderson, R.: Comparison of optimal performance at 300kev of three direct electron detectors for use in low dose electron microscopy. Ultramicroscopy
**147**, 156–163 (2014)View ArticleGoogle Scholar - Gautam, A., Ophus, C., Lançon, F., Denes, P., Dahmen, U.: Analysis of grain boundary dynamics using event detection and cumulative averaging. Ultramicroscopy
**151**, 78–84 (2015)View ArticleGoogle Scholar - Park, J., Elmlund, H., Ercius, P., Yuk, J.M., Limmer, D.T., Chen, Q., Kim, K., Han, S.H., Weitz, D.A., Zettl, A., et al.: 3D structure of individual nanocrystals in solution by electron microscopy. Science
**349**(6245), 290–295 (2015)View ArticleGoogle Scholar - Tate, M.W., Purohit, P., Chamberlain, D., Nguyen, K.X., Hovden, R., Chang, C.S., Deb, P., Turgut, E., Heron, J.T., Schlom, D.G., et al.: High dynamic range pixel array detector for scanning transmission electron microscopy. Microsc. Microanal.
**22**(01), 237–249 (2016)View ArticleGoogle Scholar - Li, X., Mooney, P., Zheng, S., Booth, C.R., Braunfeld, M.B., Gubbens, S., Agard, D.A., Cheng, Y.: Electron counting and beam-induced motion correction enable near-atomic-resolution single-particle cryo-EM. Nat. Methods
**10**(6), 584–590 (2013)View ArticleGoogle Scholar - Nogales, E.: The development of cryo-EM into a mainstream structural biology technique. Nat. Methods
**13**(1), 24–27 (2016)View ArticleGoogle Scholar - Glaeser, R.M.: How good can cryo-EM become? Nat. Methods
**13**(1), 28–32 (2016)View ArticleGoogle Scholar - Ophus, C., Ercius, P., Sarahan, M., Czarnik, C., Ciston, J.: Recording and using 4D-STEM datasets in materials science. Microsc. Microanal.
**20**(S3), 62–63 (2014)View ArticleGoogle Scholar - Ozdol, V., Gammer, C., Jin, X., Ercius, P., Ophus, C., Ciston, J., Minor, A.: Strain mapping at nanometer resolution using advanced nano-beam electron diffraction. Appl. Phys. Lett.
**106**(25), 253107 (2015)View ArticleGoogle Scholar - Pekin, T.C., Gammer, C., Ciston, J., Minor, A.M., Ophus, C.: Optimizing disk registration algorithms for nanobeam electron diffraction strain mapping. Ultramicroscopy
**176**, 170–176 (2017)View ArticleGoogle Scholar - Panova, O., Chen, X.C., Bustillo, K.C., Ophus, C., Bhatt, M.P., Balsara, N., Minor, A.M.: Orientation mapping of semicrystalline polymers using scanning electron nanobeam diffraction. Micron
**88**, 30–36 (2016)View ArticleGoogle Scholar - Shibata, N., Findlay, S.D., Kohno, Y., Sawada, H., Kondo, Y., Ikuhara, Y.: Differential phase-contrast microscopy at atomic resolution. Nat. Phys.
**8**(8), 611–615 (2012)View ArticleGoogle Scholar - Ophus, C., Ciston, J., Pierce, J., Harvey, T.R., Chess, J., McMorran, B.J., Czarnik, C., Rose, H.H., Ercius, P.: Efficient linear phase contrast in scanning transmission electron microscopy with matched illumination and detector interferometry. Nat. Commun.
**7**, 10719 (2016)View ArticleGoogle Scholar - Yang, H., Rutte, R., Jones, L., Simson, M., Sagawa, R., Ryll, H., Huth, M., Pennycook, T., Green, M., Soltau, H., et al.: Simultaneous atomic-resolution electron ptychography and z-contrast imaging of light and heavy elements in complex nanostructures. Nat. Commun.
**7**, 12532 (2016)View ArticleGoogle Scholar - Allen, L., Findlay, S., Oxley, M., Rossouw, C.: Lattice-resolution contrast from a focused coherent electron probe. Part I. Ultramicroscopy
**96**(1), 47–63 (2003)View ArticleGoogle Scholar - Findlay, S., Allen, L., Oxley, M., Rossouw, C.: Lattice-resolution contrast from a focused coherent electron probe. Part II. Ultramicroscopy
**96**(1), 65–81 (2003)View ArticleGoogle Scholar - Kirkland, E.: Advanced Computing in Electron Microscopy. Springer Science & Business Media, New York (2010)Google Scholar
- Bethe, H.: Theorie der beugung von elektronen an kristallen. Ann. Phys.
**392**(17), 55–129 (1928)View ArticleGoogle Scholar - Zuo, J., Spence, J.: Electron Microdiffraction. Springer Science & Business Media, New York (2013)Google Scholar
- Cowley, J.M., Moodie, A.F.: The scattering of electrons by atoms and crystals. I. A new theoretical approach. Acta Crystallogr.
**10**(10), 609–619 (1957)View ArticleGoogle Scholar - Shukla, A.K., Ramasse, Q.M., Ophus, C., Duncan, H., Hage, F., Chen, G.: Unravelling structural ambiguities in lithium-and manganese-rich transition metal oxides. Nat. Commun.
**6**, 8711 (2015)View ArticleGoogle Scholar - Van den Broek, W., Jiang, X., Koch, C.: FDES, a GPU-based multislice algorithm with increased efficiency of the computation of the projected potential. Ultramicroscopy
**158**, 89–97 (2015)View ArticleGoogle Scholar - Yankovich, A.B., Berkels, B., Dahmen, W., Binev, P., Sanchez, S.I., Bradley, S.A., Li, A., Szlufarska, I., Voyles, P.M.: Picometre-precision analysis of scanning transmission electron microscopy images of platinum nanocatalysts. Nat. Commun.
**5**, 4155 (2014)View ArticleGoogle Scholar - Yu, M., Yankovich, A.B., Kaczmarowski, A., Morgan, D., Voyles, P.M.: Integrated computational and experimental structure refinement for nanoparticles. ACS Nano
**10**(4), 4031–4038 (2016)View ArticleGoogle Scholar - Kim, H., Zhang, J.Y., Raghavan, S., Stemmer, S.: Direct observation of Sr vacancies in SrTiO\(_3\) by quantitative scanning transmission electron microscopy. Phys. Rev. X
**6**(4), 041063 (2016)Google Scholar - Xu, R., Chen, C.-C., Wu, L., Scott, M., Theis, W., Ophus, C., Bartels, M., Yang, Y., Ramezani-Dakhel, H., Sawaya, M.R., et al.: Three-dimensional coordinates of individual atoms in materials revealed by electron tomography. Nat. Mater.
**14**(11), 1099–1103 (2015)View ArticleGoogle Scholar - Yang, Y., Chen, C.-C., Scott, M., Ophus, C., Xu, R., Pryor Jr., A., Wu, L., Sun, F., Theis, W., Zhou, J., Eisenbach, M., Kent, P.R., Sabirianov, R.F., Zeng, H., Ercius, P., Miao, J.: Deciphering chemical order/disorder and material properties at the single-atom level. Nature
**542**, 75–79 (2017)View ArticleGoogle Scholar - Johnson, J.M., Im, S., Windl, W., Hwang, J.: Three-dimensional imaging of individual point defects using selective detection angles in annular dark field scanning transmission electron microscopy. Ultramicroscopy
**172**, 17–29 (2017)View ArticleGoogle Scholar - Barthel, J.: Time-efficient frozen phonon multislice calculations for image simulations in high-resolution STEM. Proc. 15 th Euro. Microsc. Cong. 744 (2012). http://www.emc2012.org.uk/documents/Abstracts/Abstracts/EMC2012_0744.pdf
- Grillo, V., Rotunno, E.: STEM_CELL: a software tool for electron microscopy: Part I—simulations. Ultramicroscopy
**125**, 97–111 (2013)View ArticleGoogle Scholar - Allen, L., D’Alfonso, A.J., Findlay, S.: Modelling the inelastic scattering of fast electrons. Ultramicroscopy
**151**, 11–22 (2015)View ArticleGoogle Scholar - Hosokawa, F., Shinkawa, T., Arai, Y., Sannomiya, T.: Benchmark test of accelerated multi-slice simulation by GPGPU. Ultramicroscopy
**158**, 56–64 (2015)View ArticleGoogle Scholar - Lobato, I., Van Aert, S., Verbeeck, J.: Progress and new advances in simulating electron microscopy datasets using MULTEM. Ultramicroscopy
**168**, 17–27 (2016)View ArticleGoogle Scholar - Kirkland, E.J.: Computation in electron microscopy. Acta Crystallogr. Sect. A Found. Adv.
**72**(1), 1–27 (2016)View ArticleGoogle Scholar - Chen, J., Van Dyck, D., de Beeck, M.O., Broeckx, J., Van Landuyt, J.: Modification of the multislice method for calculating coherent STEM images. phys. Stat. Sol. (A)
**150**(1), 13–22 (1995)View ArticleGoogle Scholar - Chen, C.-C., Zhu, C., White, E.R., Chiu, C.-Y., Scott, M., Regan, B., Marks, L.D., Huang, Y., Miao, J.: Three-dimensional imaging of dislocations in a nanoparticle at atomic resolution. Nature
**496**(7443), 74–77 (2013)View ArticleGoogle Scholar - Ricolleau, C., Le Bouar, Y., Amara, H., Landon-Cardinal, O., Alloyeau, D.: Random vs realistic amorphous carbon models for high resolution microscopy and electron diffraction. J. Appl. Phys.
**114**(21), 213504 (2013)View ArticleGoogle Scholar - Pennycook, S.J., Nellist, P.: Scanning Transmission Electron Microscopy: Imaging and Analysis. Springer, Berlin (2011)View ArticleGoogle Scholar
- Jones, P., Rackham, G., Steeds, J.: Higher order laue zone effects in electron diffraction and their use in lattice parameter determination. Proc. Royal Soc. Lond. A Math. Phys. Eng. Sci.
**354**, 197–222 (1977)View ArticleGoogle Scholar